This application protocol describes the preparation of tetramer solutions, staining, and sorting of SARS-CoV-2–specific B cells using recombinant SARS-CoV-2 spike proteins and the MACSQuant Tyto Cell Sorter.
The following protocol is an example when using Streptavidin, PE and Streptavidin, PE-Vio 770. Other streptavidin conjugates can be used as well. Always check the concentration of the stock beforehand and adapt the protocol accordingly.
For the preparation of spike-tetramer solutions, the Recombinant SARS-CoV-2 Spike-Prot (HEK)-Biotin was conjugated to Streptavidin, PE and Strepdavidin, PE-Vio 770 with a ratio of 4 mol to 1 mol, respectively (refer to table 1).
▲ Note: Always prepare spike-tetramer solutions fresh before the use. They can be stored for up to 24 hours at 2–8 °C, protected from light.
Each test indicated in table 1 corresponds to 5×106 B cells (or 107 PBMCs). When working with less than 5×106 B cells (or 107 PBMCs), use the same volumes as indicated for 1 test. When working with more than 5×106 B cells (or 107 PBMCs), scale up all reagent volumes accordingly.
Table 1: Volumes of spike-tetramer solutions
|No. of tests||Recombinant SARS-CoV-2 Spike-Prot (HEK)-Biotin (0.1 mg/mL)||Streptavidin, PE (50 µg/mL)||MACSQuant Tyto Running Buffer|
|1||3 µL||0.6 µL||1.4 µL|
|5||15 µL||3 µL||7 µL|
|10||30 µL||6 µL||14 µL|
|20||60 µL||12 µL||28 µL|
|No. of tests||Recombinant SARS-CoV-2 Spike-Prot (HEK)-Biotin (0.1 mg/mL)||Streptavidin, PE-Vio 770 (50 µg/mL)||MACSQuant Tyto Running Buffer|
|1||6 µL||1.2 µL||2.8 µL|
|5||30 µL||6 µL||14 µL|
|10||60 µL||12 µL||28 µL|
|20||120 µL||24 µL||56 µL|
▲ Note: If using an IgG Antibody, the other antibodies should not be IgG as isotype. Therefore, the use of REAfinity™ Antibodies for this panel design is not recommended, as REAfinity Antibodies are all IgG isotype.
For each test (5×106 B cells or 107 PBMCs) mix 2 μL of each fluorochrome-conjugated antibody, 5 μL of 7-AAD Staining Solution, 5 μL of spike‑tetramer-PE, and 10 μL of spike-tetramer-PE-Vio 770.
Fill up to 100 μL with the MACSQuant Tyto Running Buffer.
▲Note: Try different volumes of tetramers conjugates in order to achieve the best double staining results (refer to section flow cytometry analysis).
Suggested antibody panel
|V1||IgG Antibody, anti-human, VioBlue®||IS11-3B2.2.3|
|V2||IgA Antibody, anti-human, VioGreen™||IS11-8E10|
|B1||CD27 Antibody, anti-human, Vio Bright FITC||M-T271|
|B3||7-AAD Staining Solution|
|R1||IgM Antibody, anti-human, APC||PJ2-22H3|
|R2||CD19 Antibody, anti-human, APC-Vio 770||LT19|
B cells can be enriched from single-cell suspension of peripheral blood mononuclear cells (PBMCs) using the REAlease® CD19 MicroBead Kit (# 130-117-034) accordingly to the data sheet including removal of magnetic labeling.
SARS-CoV-2 Spike B Cell Analysis Kit, human (# 130-128-022)
7-AAD Staining Solution (# 130-111-568)
IgA Antibody, anti-human, VioGreen™ (# 130-113-481)
IgG Antibody, anti-human, VioBlue® (# 130-119-881)
IgM Antibody, anti-human, APC (# 130-122-915)
CD19 Antibody, anti-human, APC-Vio 770 (# 130-113-166)
CD27 Antibody, anti-human, Vio Bright FITC (# 130-113-634)
MACSQuant® Analyzer 10 (# 130-096-343) or other flow cytometers equipped with violet (405 nm) and red (635 nm) lasers able to discriminate FITC, PE, and APC fluorescence.
MACS Chill 96 Rack (# 130-094-459) when using MACSQuant Analyzer 10
MACSQuant Calibration Beads (# 130-093-607) when using MACSQuant Analyzer 10
MACS Comp Bead Kits for optimal compensation of the fluorescence spillover from fluorochrome-conjugated antibodies
Unless otherwise specifically indicated, Miltenyi Biotec products and services are for research use only and not for therapeutic or diagnostic use.
If not otherwise specified, all marks displayed on the Miltenyi Biotec Internet site and documents are subject to the trademark rights of Miltenyi Biotec, including each of Miltenyi Biotec's primary brands, its model name plates, and its corporate logos and emblems.
For copyright note and a complete list of trademarks visit www.miltenyibiotec.com/legalnotes.